ATC Portable Digital PH Tester
- Range: 0.0 to 14.0 pH;
- Resolution: 0.1 pH;
- Accuracy: ±0.1 pH;
- Calibration: Manual, 1 point;
- Temperature Compensation Automatic 0 to 50°C;
- Environment: 0 to 50°C (32 to 122°F);
- RH max 95%
- Batteries Type: 4 x 1.5V alkaline;
- Batteries Life: approx. 150 hours of use;
- Remove the protective cap. Do not be alarmed if white crystal, appear around the cap. This is normal with pH electrodes and they dissolve when rinsed with water.
- Turn the meter on sliding the switch on the top.
- Immerse into solution up to the max immersion level.
- Stir gently and wait until the display stabilizes.
- After use, switch off pH meter, use the water clean the electrode and replace the protective cap.
DO NOT USE DISTILLED OR DEIONIZED WATER FOR STORAGE PURPOSES.
Large fluctuations in readings could be due to lack of calibration, dry electrode or rundown batteries.
- Immerse the tester up to the maximum immersion level in pH 6.86 buffer solution;
- Allow the reading to stabilize and using a small screwdriver adjust the calibration trimmer to read 6.9;
- Calibration is now complete.
ALWAYS USE FRESH BUFFERS FOR CALIBRATION.
When the meter cannot be switched on or the display fades, pull out the battery compartment and replace all three 1.5V batteries, while paying attention to their polarity. Batteries should only be replaced in a safe area using the battery type specified in this instruction manual.
- 1x ATC Portable Digital PH Tester
- 1 x Mini-screw Driver
- 1 x Instruction Sheet
- 1 x A Bag of Mixed Phosphate
- 1x a bag of Potassium Hydrogen Phthalate
PH- TESTER (DIGITAAL)
- Model: atc ph tester
- Beschikbaarheid: Op voorraad
- Excl. BTW: € 12,40